Pearl Trilogy Specifications
Operating Conditions: For indoor use only; operating temperature 15-35 °C and dew point < 22 °C, non-condensing; maximum operating temperature may be reduced at elevations above 2000m.
Environmental Conditions: Pollution Degree 2.
Detector Type: CCD. Thermoelectrically cooled.
Acquisition Time: 500 ms to 40 s.
Acquisition Time: 500 ms to 60 s.
Wavelength Maxima for Excitation/Emission Bands:
|700 Channel||685 nm||720 nm|
|800 Channel||785 nm||820 nm|
|Bioluminescence||NA||450 - 650 nm|
Dyes: Optimized for LI‑COR® IRDye® 680RD and IRDye 800CW.
Resolution: 85, 170, and 255 micron.
Image Display Options: Pseudocolor, grayscale, single color (red, green, or blue), or two colors with overlapping fluorescence displayed in a third color.
Imaging Drawer: 16.8 cm W × 12 cm D (6.6” W × 4.75” D). Vertical clearance to top of drawer is 7.5 cm (3”).
Field of View: 11.2 cm W × 8.4 cm D (4.4” W × 3.3” D) at the surface of the imaging
Power Requirements: Universal input range is between 100-127 VAC and 200-240 VAC (voltage fluctuations not to exceed 10% of the nominal voltage); 50/60 Hz; at 120V, 4 Amp maximum; at 120V, 0.3 Amp typical when idle. Overvoltage Category II.
Computer Connection Cable: Cat. 5e RJ45, 10BASE-T/100BASE-TX/1000BASE-T. Use only the supplied cable.
Size: 41 W × 41 D × 66 cm H (16” W × 16” D × 26” H). Depth with imaging drawer open is 63.5 cm (25”).
Weight: 25 kg (55 lb).
The Pearl Imager is certified as a Class 1 laser product. Hazardous laser radiation is not emitted outside the instrument. Radiation emitted inside the Pearl Imager is confined within protective housings and external covers. A series of safety interlocks ensures that laser beams cannot escape during any phase of user operation.
Embedded Firmware Source Code
The embedded firmware running on the